In pipeline assembly, one piece backing flanges are used with a loose hinged insert. The following couplings are available:-
This is the most frequently used coupling since it is corrosion-resistant, light weight and does not require electrostatic earthing. The plastic material is solvent and acid resistant glass fibre-loaded phenolic resin.
This coupling has the above features with the advantage that the flanges ca7005-3, n be connected directly to other materials of construction with flanges to ISO 7005-2, PN 10, ISO 7005-3, PN 10 and DIN 2501, PN 10.
Couplings with Silumin backing flanges. These couplings are used for high temperature duties for example where lagging is fitted over the top of the coupling or where certain corrosive substances are being processed.
All the above couplings are suitable for operation at 200 deg C on product side and for the full range of operating pressures. The plastic flanges are not to be used in duties where we shall be using lagging over the coupling. The standard bolts, nuts and washers are bright zinc-plated. Stainless steel bolting is available on request.
When processing the fluids which tend to crystallise or are particularly searching. It is advisable to use longer bolts fitted with compression springs. This provides additional elasticity in the joint and ensures permanent leak tightness and freedom from maintenance under extreme conditions.
PTFE is the normal gasket material due to its universal corrosion-resistance.
Some Of The Precautions To Avoid Tensile Stress In Glass Equipment:-
After initial assembly, the following procedure us necessary to ensure a leak-free plant:
The glass plant can then be checked for leak tightness by filling with water. This is the method used for the vast majority of new installations. The hydrostatic head of water MUST NOT exceed the maximum allowable operating pressure.
If gas testing is called for, the test pressure should be kept as low as possible. A further option for tightness testing is to apply a vacuum. A halogen leak probe can be used to detect leaks or the rise in pressure over a period of time can be measured and compared to the calculated figure. The figure for joints with a ground face and PTFE gaskets is 1 X 10-2 mbar l/s per meter length of gasket. For example a DN100 joint will have a leakage rate of 5 X 10-4 mbar l/s.
In case of couplings fitted with compression springs, it is essential to use a torque spanner to tighten the bolts. Do not exceed the maximum torque loading. It is not necessary to heat the entire plant and re-torque the bolts when springs have been fitted because the springs will take up any initial relaxation of the PTFE gasket.
Process commissioning, as with design and installation, is a key component in the range of service offered by ATR. ATR can provide fitters and engineers for commissioning as well as for installation and leak testing.